When running Bluetooth® LE tests, I get a “Bluetooth® TX LE retrieve analysis results not valid” error message from IQfact+. What does this mean, and how can I fix it?
The capture signal was not able to be analyzed for the measurement that was requested. A couple of likely causes:
- Signal level too high: It’s possible the VSA was saturated. Increase the RLEV.
- Capture too short: The analysis needs sufficient data to perform a measurement. Try increasing capture length.
- Incorrect signal for the analysis: Make sure the signal you are measuring is a valid signal for the analysis type. For instance, you can’t run Bluetooth® Low Energy analyses on a Bluetooth® Classic signal.